The Accu-Chek Aviva was first in an evaluation of a dozen meters conducted by a team of testing experts at Germany’s University of Ulm led by Guido Freckmann, M.D. It led the field in accuracy and precision.
Few studies of meter accuracy have appeared in the more than 20 years that I have been writing about diabetes. Dr. Freckmann and his team of researchers have been the most relevant, reliable, and prolific in testing our meters. But some of the meters that they evaluated aren’t available in the United States.
The leading diabetes journal that evaluates our meters just released the full text of this meter accuracy study. The editors tell me that the study will be free online only until March 31.
Most Relevant Info
So don’t wait too long if you want to review the whole thing. Meanwhile, I summarize below what I think is the most relevant information.
The journal is Diabetes Technology & Therapeutics, and the study is “Evaluation of 12 Blood Glucose Monitoring Systems for Self-Testing: System Accuracy and Measurement Reproducibility.” The journal published it two years ago, but until now only the abstract has been freely available to us.
Dr. Freckmann and his colleagues rated the meters against both the current and proposed standards. The current standard can be met more easily. It requires that at least 95 percent of the results fall within plus or minus 15 mg/dl at blood glucose levels below 75 mg/dl and within plus or minus 20 mg/dl at levels greater than or equal to 75 mg/dl. The International Organization for Standardization (ISO) set this level back in 2003, and it is the standard not only in Europe but also in the United States because the U.S. Food and Drug Administration uses it.
The Tighter Standard
ISO’s 2013 standard is tighter, but is still pending approval both in the U.S. and Europe. To be acceptable at least 95 percent of the results have to be within plus or minus 15 mg/dl at blood glucose levels below 100 mg/dl and within plus or minus 15 percent at blood glucose levels greater than or equal to 100 mg/dl.
Ten of the 12 meters that Dr. Freckmann and his team evaluated met the higher standard. Two meters, which they didn’t name, failed the test. The FDA hasn’t approved six others for sale in the United States.
But four of the five top meters are available in the United States. While the Accu-Chek Aviva got the best score in this evaluation, the Contour XT, sold here as the Contour Next EZ, is the next most accurate and precise meter tested that is available here. Numbers four and five are the GE100 and GE200, also available here.
Dr. Bernstein’s Recommendation
Of course, I am aware that Richard K. Bernstein, M.D., recommends the Abbott FreeStyle Freedom Lite meter as “the most accurate.” He is the author of Dr. Bernstein’s Diabetes Solution and the leading exponent of the very low-carb diet for people with diabetes.
In fact, I tested the Abbott FreeStyle Freedom Lite when it first came on the market. My results were so variable that I immediately threw it away. Now that I’ve read the results of the evaluation by Dr. Freckmann’s team, I’m glad that I still test with one of the Accu-Chek Aviva meters that I started using more than 10 years ago.
This article is based on an earlier version of my article published by HealthCentral.
Never Miss An Update
Subscribe to my free newsletter “Diabetes Update”
I send out my newsletter on first of every month. It covers new articles and columns that I have written and important developments in diabetes generally that you may have missed.